Product details of OH UA6013L Auto Range Digital LCD Capacitor Capacitance Test Tester
6013L type digital capacitance meter is pocket, convenient
operation, accurate numerical reading, low-consumption, driven by
battery, and showing numbers in a big LCD. Being accurate in
measuring, it covers 9 measuring sections, a wide rang from 0.1pF
to 20,000ÔøΩÔøΩF. It can be used in testing error testing,
numerical analysis, selecting capacitance, measuring unknown
capacitance, matching capacitance, and measuring the capacitance of
cable, switch and printing electrocircuit.
It is applied with CMOS double-bevel A/D convertor that is
automatic in zeroing and polar selection, and makes instruction for
beyond measuring range.
Wide measuring range, covering 9 measuring sections from 0.1pF to
20,000ÔøΩÔøΩF that includes nominal value of any capacitance.
Biggest display value: 1999(3 1/2).
Reading: 2-3 readings/sec.
Zeroing: There is a zeroing knob on the front, easy for
Data hold: Put the ÔøΩÔøΩHOLDÔøΩÔøΩ switch to 'ON' and the DH sign
will appear on the display.
Temperature for accurate measuring: 25ÔøΩÔøΩ5ÔøΩÔøΩC
Temperature range: working temperature: 0 to 40ÔøΩÔøΩC
Storage temperature: -10to 50ÔøΩÔøΩC
Relative humidity: 80%
Power supply:9V laminated battery(Not included)
Accuracy: ÔøΩÔøΩ(%readingÔøΩÔøΩdigits) Quality guarantee: one
Environment temperature: 25ÔøΩÔøΩÔøΩ5ÔøΩÔøΩ
Relative humidity: 80%
Rang Accuracy Definition Testing frequency
200pF ÔøΩÔøΩ0.5% ÔøΩÔøΩ20digit 0.1pF 800Hz
2nF ÔøΩÔøΩ0.5%ÔøΩÔøΩ20digit 1pF 800Hz
20nF ÔøΩÔøΩ0.5%ÔøΩÔøΩ20digit 10pF 800Hz
200nF ÔøΩÔøΩ0.5%ÔøΩÔøΩ20digit 0.1ÔøΩÔøΩF 800Hz
2ÔøΩÔøΩF ÔøΩÔøΩ0.5%ÔøΩÔøΩ20digit 1nF 800Hz
20ÔøΩÔøΩF ÔøΩÔøΩ0.5%ÔøΩÔøΩ20digit 10nF 80Hz
200ÔøΩÔøΩF ÔøΩÔøΩ0.5%ÔøΩÔøΩ20digit 0.1ÔøΩÔøΩF 8Hz
2000ÔøΩÔøΩF ÔøΩÔøΩ2.0%ÔøΩÔøΩ20digit 1ÔøΩÔøΩF 8Hz
20000ÔøΩÔøΩF ÔøΩÔøΩ4.0%ÔøΩÔøΩ20digit 10ÔøΩÔøΩF 8Hz
Dimension:148 x 75 x 33 mm
Package weight:193 g
1 x Capacitance Tester
2 x Probes
Firstly, check a 9V battery whether its electric voltage is short
that will show signal on the left-upper of the LCD. Secondly, pay
attention to the polarity or the capacitance to be tested. Thirdly,
recharge the capacitance before measuring.
1. Select proper measuring section according to the capacitance to
2. Check and use the zeroing knob on the front of the meter to
adjust the value to zero before measuring.
3. Insert the black sensor into the jack with ÔøΩÔøΩ-ÔøΩÔøΩsignal
and the red sensor into the jack with ÔøΩÔøΩ+ÔøΩÔøΩ.
4. Connect the capacitance to be tested with the two sensors. Pay
attention to the polarity of the capacitance ! Read capacitance
numerical value displayed.
a. If you don't know the approximate value of a capacitance, you
should select measuring section of 200nF and then make relevant
adjusting according to practical situation.
b. If the capacitance is in short circuit, the meter shows 'beyond
measuring range'. If the capacitance is in electric leakage, the
meter shows 'beyond measuring range' or 'much bigger than the
normal value'. If the capacitance is open-circuit, the meter shows
'Zero' or 'several pF '.
c. In testing small capacitance, directly insert the capacitance
into the jack instead of using the sensors for better accuracy.
The meter is of highly accurate electronic apparatus. Don't change
its internal circuit rashly for no damage to the meter.
1. Don't exert electric voltage upon the jack parts of the meter
for no damage to the meter.
2. Don't test the sensors in short circuit , or the battery will be
consumed out quickly.
3. Don't use a meter whose back cover doesn't cover the meter
4. Always replace battery and fuse after taking sensors out of a
meter that is switched off. First, screw bolts in the back cover
out. Second, take off the back cover. Third, replace battery and
fuse according to specification provisioned in the operation
manual. Suitable fuse for this meter is 200mA/250V, which external
dimension is ÔøΩÔøΩ5ÔøΩÔøΩ20ÔøΩL.
5. Please take out the battery when not using for a long time.